The angles of the guide posts were made semi-parallel to the adjacentnatural teeth. However, the angles are not too critical for theDIAGNOSTIC guide, as they most likely will need to be corrected.
Occlusal view Facial view3 X 8 mm guide sleeves are placed over the guide posts.
Undercuts were blocked out and petroleum jelly was used to lubricate the cast.Triad® Gel was added to the brackets of the guide sleeves and the adjacentteeth to form the body of the DIAGNOSTIC guide.
Facial view of fabricated template.The DIAGNOSTIC guide is now ready to beplaced in the patient for a Cone Beam X-ray.
Preview of Pre-op Corrections #23 LINEAR CORRECTION 10° correction toward the buccal #27 LINEAR CORRECTION 6° correction toward the lingual
# 23 Initial Alignment Software Evaluation axial cross sectional Invivo5 Invivo5 Anatomage Anatomage tangential volumetricNOTE Each time a change is made in the software go back to theinitial alignment view and calculate the angle in the 2nd plane.
Invivo5Anatomage #23 CLOSE UP with software grid tangential view cross sectional view requires no correction requires a 10° correction toward the distal toward the buccal
#27 Axial view should always be adjusted to the 90° cross sectional plane of the alveolar ridge. axial view occlusal view Bucco-lingual plane Invivo5Anatomage
#27 CLOSE UP LINEAR & ANGULAR Evaluation tangential view cross sectional view guide sleeve requires no correction guide sleeve requires a 6° away from the adjacent tooth correction toward the lingual (left)Invivo5Anatomage
Preview of Pre-op Corrections #27 LINEAR CORRECTION: 1.0 mm to the buccal ANGULAR CORRECTION 6° bend BEND to the lingual. #23 LINEAR CORRECTION 0.5 mm to the lingual
Occlusal view Buccal view#27 correction: 1.0 mm offset guide post to the buccal & bent 6° to the lingual.#23 correction: 0.5 mm offset guide post to the lingual / no change in the angle
Occlusal view Buccal view3.3 mm open guide sleeves placed on offset magnetic guide posts.
Guide Right™ Products shown in this case 1 - 3/32” Drill 2 - 3 mm Guide Post 2 - 3 mm cylindrical Guide Sleeve 1 - 3 mm x 0.5 mm Offset Magnetic Guide Post 1 - 3 mm x 1.0 mm Offset Magnetic Guide Post 2 - 3.3 mm Open Guide Sleeves Generation ll ▪ Guide Post Bending Tool Triad® Gel 1.800.314.0065 • www.deplaque.com
DéPlaqueGuide Right™ Surgical Guide System fabricate ▪ evaluate ▪ correct ▪ verify ▪ placeStart With Precision. Place With Confidence.™ 1.800.314.0065 • www.deplaque.com