Undercuts blocked out.30 mm guide posts placed in 3/32” holes.
3 mm guide sleeves placed on 3 x 30 mm guide posts with cleats toward the palatal.
Triad® Gel added to guide sleeves capturing the cleats and adjacent teeth to form the DIAGNOSTIC guide
Preview of Pre-op Corrections #12 LINEAR CORRECTION1st : 1 mm toward the buccal2nd : 1 mm distally using a 1.5 mm offset guide post rotated 45 ° #14 LINEAR CORRECTION 1 mm toward mesial using1 mm offset guide post Cosine is applied to determine correction data. * see next 3 slide
DéPlaque Pythagorean TheoremCOSINE Calculations for Guide Right™ Guides The cosine of 45° is recommended for Guide Right corrections BECAUSE rotating the offset guide post half way between 2 adjacent planes (90° apart) is 45°.
Correction based on the calculations from the Pythagorean TheoremTo move the position of the guide sleeve 1.4 mm both mesially & buccally: ► use a 3 mm X 1.5 mm offset guide post ► and direct the offset 45º facially and buccally. Cosine: 1.5 mm X 0.71 = 1.06 mm see Slideshow > Use of Pythagorean Theorem in #9 Single Implant Case
A= cosine of 45º X 1.5(0.707 X 1.5 mm = 1.06 mm
LINEAR CORRECTIONIndicates guide sleeve should be moved 1 mm distally # 12 occlusal view cross sectional view Invivo5 Anatomage Invivo5 Anatomage volumetric tangential view
# 12 axial view: wrong plane Axial view should always be adjusted to the 90° cross sectional plane of the alveolar ridge.Invivo5Anatomage
Invivo5 #12 axial plane rotated to about 90°Anatomage
# 12 cross sectional view tangential viewInvivo5Anatomage by rotating a 1.5 mm offset guide post 45 ° The guide sleeve is moved 1 mm toward the buccal and 1 mm to the distal(Pythagorean Theorem)
# 14 LINEAR CORRECTIONmove guide sleeve 1 mm mesially axial view cross sectional view Invivo5 Anatomage tangential view volumetric
#14Invivo5Anatomage Axial view should always be adjusted to the 90° cross sectional plane of the alveolar ridge.
tangential #14 moving the position of the guide sleeve 1 mm mesiallyInvivo5 will require a 1 mm offset guide postAnatomage
Understanding Invivo5 Grid Measurements SOFTWARE GRID each square: 5 X 5 mm each side of the square: 5 mm center mark is 2.5 mm tangential
#14To calculate the 1 mm linear movement ofthe guide sleeve► The distance from the orange linedrawn through the long axis of the virtualimplant to the green line drawn throughthe center axis of the guide sleeve ismeasured. tangential Invivo5 Anatomage
Review of Pre-op Corrections #12 LINEAR CORRECTION1st : 1 mm toward the buccal2nd : 1 mm distally using a 1.5 mm offset guide post rotated 45 ° #14 LINEAR CORRECTION 1 mm toward the mesial using1 mm offset guide post Cosine is applied to determine correction data. * see next 3 slide
Magnetic Guide Posts Straight 3.0 mm od Offset 3.0 od X 0.5 mm os 3.0 od X 1.0 mm os 3.0 od X 1.5 mm os 3.0 od X 2.0 mm os 3.0 od X 2.5 mm os 3.0 od X 3.0 mm os
Open Magnetic Guide Sleeve sets 3.0 mm x 5.0 mm Open Guide Sleeve (short) 3.0 mm x 8.0 mm Open Guide Sleeve (long) 3.0 mm Magnetic Guide Post Straight 3.0 mm x 1.0 mm Offset Magnetic Guide Post 2.0 mm x 8.5 mm Pilot Drills 2.0 mm x10.0 mm Pilot Drill 4.0 mm x 5.0 mm Open Guide Sleeve (short) 4.0 mm x 8.0 mm Open Guide Sleeve (long) 4.0 mm Magnetic Guide Posts Straight 4.0 mm x 1.0 mm Offset Magnetic Guide Post 2.0 x 5.0 mm Pilot Drill 2.0 x 10.0 mm Pilot Drill
Guide Right™ Products shown in this case 1 - 3/32” Drill 2 - 3.0 mm x 5.0 mm Open Guide Sleeve (short) 2 - 3.0 mm Guide Sleeve 1 - 3 mm X 1.5 mm Offset Magnetic Guide Posts 1 - 3 mm X 1.0 mm Offset Magnetic Guide Posts 1 - 4 mm od. Spacer (fits over the 3 mm guide post) 1 - 3 mm X 0.5 mm Magnetic Offset Guide Post 2 - 3 mm Open Guide Sleeve Guide Post Bending Tool Triad® Gel 1.800.314.0065 • www.deplaque.com
Guide Right™ GUIDE POST BENDING TOOL SINGLE BEND reviewStep 1 Place bending tool plate on a secure flat surface with the degree increments at the top & the stainless steel bar with the v-cut at the bottom.Step 2 Locate 3/32” hole in the center of the v-cut and place the bottom half of the guide post into the hole. Tighten the set screw.Step 3 Locate the hole in the bottom of the stylus that you will use that will fit over the top half of the guide post (3.0 mm, 4.0 mm or 5.0 mm).Step 4 Fit the stylus over the guide post securely with the point directed at zero degrees and the bottom of the stylus in contact with the V block.Step 5 Using the stylus as a lever, bend the guide post to the degree of angle of correction. You may need to ease the point of the stylus beyond the point of the desired degree.Step 6 Loosen screw and remove guide post and the stylus to find the guide post bent to the desired angle.
Guide Right™ GUIDE POST BENDING TOOL COMPOUND BEND overviewStep 1 Position a straight or offset guide post in the bending plate, tightening the set screw against one of the flat surfaces on the lower half of the guide post.Step 2 The 1st bend can be made to the right or left direction.Step 3 The set screw is loosened and the guide post is rotated 90 ° next flat surface.Step 4 The 2nd bend in the second plane is made after rotating the guide post up away from the surface of the bending plate to register the stylus point back at 0 degrees.Step 5 Slide the stylus support bar down under the stylus until it supports the stylus. Tighten the side screws before making the second bend.Step 6 The second bend can be made in either direction according to the x-ray.Step 7 Remove the stylus and place the guide post back in the cast with the appropriate side indicated by a mark facing the buccal or lingual surface. Be sure the post is in the correct position. If the post needs to be corrected by a linear movement an offset guide post can be used. Off sets available in the 3 mm guide post: 0.5,1,1.5, 2.0 ,or 3.0 mm.
A System of Componentsfor the fabrication and correction of diagnostic & surgical guides in one or two dimensions In-office or lab fabrication Evaluate with 2D & 3D imaging Allows linear and angular correction Enables precision implant placement Cost effective 1.800.314.0065 • www.deplaque.com
DéPlaqueGuide Right™ Surgical Guide System fabricate ▪ evaluate ▪ correct ▪ verify ▪ place Start With Precision. Place With Confidence.™ 1.800.314.0065 • www.deplaque.com