Guide Right™ Fibular Graft Case Featuring: # 29 and #30 3 mm Guide Posts 3 mm Diagnostic Guide Sleeves 3 mm Magnetic Guide Posts 3.3 and 4 mm Open Guide Sleeves Generation ll / Guide Post Bending Tool Invivo5 Software Evaluation AnatomageKARLS
This patient had a section of the mandible removed & graftedwith a vascular fibular graft 2 years prior to implant placement.
# 30 Shows initial alignment is NOT cross sectional GOOD cross sectional view Invivo5Anatomage
# 304.8 mm X 10 mm Straumann Implant (virtual image)1 mm offset guide post > 15° bend toward lingualaxial cross sectional Invivo5 Anatomagetangential volumetric
# 30 shows a 15° angular correction is necessary for the guide sleeve to line up with the position of thecross sectional virtual implant tangential view Invivo5Anatomage
Guide Right™ Generation ll ▪ Guide Post Bending Tool 10 ° example For more information on use of the bending tool go towww.deplaque.com ▪ Guide Right™ Generation ll Bending Tool
Preview of Pre-op Corrections #29 #30 LINEAR CORRECTION LINEAR CORRECTION 1 mm offset 1 mm offset ANGULAR CORRECTION ANGULAR CORRECTION1ST BEND: 8° toward the mesial 1 BEND: 15° toward lingual2nd BEND: 5° toward the lingual Cosine is applied to determine correction data. * see next 3 slide
# 29 Guide post positioned in the bending tool with offset positioned forward.
Magnetic Offset Guide Posts Straight 3.0 mm od Offset 3.0 od X 0.5 mm os 3.0 od X 1.0 mm os 3.0 od X 1.5 mm os 3.0 od X 2.0 mm os 3.0 od X 2.5 mm os 3.0 od X 3.0 mm os
Guide Right™Products shown in this case 3/32” Drill 3 mm Guide Posts 3 mm Diagnostic Guide Sleeves 3 mm Magnetic guide posts 3.3 mm Open Guide sleeves 4 mm Open Guide Sleeves 1.0 mm Offset Magnetic Guide Sleeve Triad® Gel 1.800.314.0065 • www.deplaque.com
DéPlaqueGuide Right™ Surgical Guide System fabricate ▪ evaluate ▪ correct ▪ verify ▪ placeStart With Precision. Place With Confidence.™ 1.800.314.0065 • www.deplaque.com