
Be the first to like this
Slideshare uses cookies to improve functionality and performance, and to provide you with relevant advertising. If you continue browsing the site, you agree to the use of cookies on this website. See our User Agreement and Privacy Policy.
Slideshare uses cookies to improve functionality and performance, and to provide you with relevant advertising. If you continue browsing the site, you agree to the use of cookies on this website. See our Privacy Policy and User Agreement for details.
Published on
To determine the accuracy and precision of pupil measurements made with the Tobii 1750 remote video eye tracker, we performed a formal metrological study with respect to a calibrated reference instrument, a medical pupillometer. We found that the eye tracker measures mean binocular pupil diameter with precision 0.10 mm and mean binocular pupil
dilations with precision 0.15 mm.
Clipping is a handy way to collect and organize the most important slides from a presentation. You can keep your great finds in clipboards organized around topics.
Be the first to comment