If the implant is placed with the guide sleeve in this position the facial wall of the bone will be violated. Therefore the axis of the guide sleeve needs to be changed
The virtual implant is repositioned in the center of the alveolar bone and a line is drawn through the long axis of the implant and the long axis of the guide sleeve witih the angle tool it is measured at 15 degrees indicating the angle correction that is made by bending the guide post 25 degrees to the buccal: The tangential view indicates that no correction is needed.
One method of checking the bend is to place the bent guide post over a print out of the view observed in the software.
This means that a 15 degree angled abutment will be needed to correct the labial direction for the temporary and the final restorations
NOTE: # 11 Initial AlignmentEach time a change is made in the software go back to theinitial line up view and calculate the angle in the 2nd plane.axial cross sectional Invivo5 Anatomage Invivo5 Anatomagetangential view volumetric
# 11 Initial Alignment of guide sleeve with virtual implantInvivo5 cross sectional viewAnatomage
Understanding Invivo5 Grid Measurements SOFTWARE GRID each square: 5 X 5 mm each side of the square: 5 mm center mark is 2.5 mm tangential
If the implant is placed with the guide sleeve in this position the facial wall of the bone will be violated.Therefore the axis of the guide sleeve needs to be changed # 11 CLOSE UP
# 11 indicationANGULAR CORRECTION: 15º bend toward the facial axial cross sectional Invivo5 Invivo5 Anatomage Anatomage tangential volumetric
# 11 CLOSE UP ANGULAR CORRECTION: 15 to the facial tangential cross sectional Invivo5Anatomage
ANGULAR CORRECTIONS are made with the Guide Right™ Generation ll Bending ToolStep by step instructions at end of slideshow
Guide Right™ Generation ll Bending Tool 10 example For more information on use of the bending tool go towww.deplaque.com ▪ Guide Right™ Generation ll Bending Tool
15 One bend to correct # 113 mm stylus placed over 3 mm guide post to make the 15 bend
# 11 Image showing 15 difference in virtual implant correction Before After Invivo5 cross sectional views
# 11 actual guide post placed on image of virtual guide post and corrected guide in place to verify 15º magnetic guide post correction One method of checking the guide post after the bend is to place the bent guide post over a print out of the view observed in the software. Invivo5Anatomage
Review of Pre-op Corrections #11 ANGULAR CORRECTION 15º bend toward the facial
Fabrication of SURGICAL Guide 3 mm Magnetic Guide Post 3 mm Open Guide Sleeve
Existing restoration removed from cast before making SURGICAL guide
Straight magnetic 3 mm guide post in planned osteotomy hole
Corrected 3 mm magnetic guide post with 15 bend toward facial
3 mm bent magnetic guide post showing 15 correction
Occlusal view of corrected 3 mm magnetic guide post and open guide sleeve
Initial Triad® Gel applied to capture the cleats ofthe open guide sleeve to form the SURGICAL guide.
Final application of gel to form the SURGICAL guide
Guide Right™Products shown in this case 1 - 3/32” drill 1 - 3 mm Guide Post 1 - 3 mm Diagnostic Guide Sleeve (1 cleat) 1 - 3 mm Magnetic Guide Post 1 - 3 mm Open Guide Sleeve Generation ll Bending Tool Triad® Gel 1.800.314.0065 • www.deplaque.com
Guide Right™ GENERATION II BENDING TOOL SINGLE BEND reviewStep 1 Place bending tool plate on a secure flat surface with the degree increments at the top & the stainless steel bar with the v-cut at the bottom.Step 2 Locate 3/32” hole in the center of the v-cut and place the bottom half of the guide post into the hole. Tighten the set screw.Step 3 Locate the hole in the bottom of the stylus that you will use that will fit over the top half of the guide post (3.0 mm, 4.0 mm or 5.0 mm).Step 4 Fit the stylus over the guide post securely with the point directed at zero degrees and the bottom of the stylus in contact with the V block.Step 5 Using the stylus as a lever, bend the guide post to the degree of angle of correction. You may need to ease the point of the stylus beyond the point of the desired degree.Step 6 Loosen screw and remove guide post and the stylus to find the guide post bent to the desired angle.
Guide Right™ GENERATION II BENDING TOOL COMPOUND BEND overviewStep 1 Position a straight or offset guide post in the bending plate, tightening the set screw against one of the flat surfaces on the lower half of the guide post.Step 2 The 1st bend can be made to the right or left direction.Step 3 The set screw is loosened and the guide post is rotated 90 ° next flat surface.Step 4 The 2nd bend in the second plane is made after rotating the guide post up away from the surface of the bending plate to register the stylus point back at 0 degrees.Step 5 Slide the stylus support bar down under the stylus until it supports the stylus. Tighten the side screws before making the second bend.Step 6 The 2nd bend can be made in either direction according to the x-ray.Step 7 Remove the stylus and place the guide post back in the cast with the appropriate side indicated by a mark facing the buccal or lingual surface. Be sure the post is in the correct position. If the post needs to be corrected by a linear movement an offset guide post can be used. Off sets available in the 3 mm guide post: 0.5,1,1.5, 2.0 ,or 3.0 mm.
A System of Componentsfor the fabrication and correction of DIAGNOSTIC & SURGICAL guides in one or two dimensions In-office or lab fabrication Evaluate with 2D & 3D imaging Allows linear and angular correction Enables precision implant placement Cost effective 1.800.314.0065 • www.deplaque.com
Guide Right™ Surgical Guide System fabricate evaluate correct verify placeStart With Precision. Place With Confidence.™ 1.800.314.0065 • www.deplaque.com