Twin-Lamp Technology• Lamp-life decrease as driven to higher energies Lamp- p g g• BBL uses two lamps supplying half of the energy needed. needed• Unlimited shots under warranty. Lamps
BBL™Large 15 X 45mm treatment area& smaller 15 X 15mm adapter> 300,000 Pulses/ dual-lamp7 FiltersIntegrated Skin CoolingHigh Power - up to 30 J/cm2Wide pulse width range - to 200msRapid TreatmentAvailability Module for Profile 110V Tabletop BBLs
Controlled Cooling By Degree• Cold sapphire crystal maintains cool treatment zone• S t System accurately sets temp. from 0ºC to 30ºC t l t t f t – Skin surface temperature changes of 4ºC can cause immediate differences in the clinical response• Temperature delivery is consistent = safety & comfort 560 nm filter 12 J/cm2 applied for 10 ms no treatment
BBL Treatment before after 1tx Photos courtesy of Michael Gold, MD
SkinTyteTM FDA Clearance• selective photocoagulation of soft tissue• with a integral thermo-electric cooler thermo-• cooling epidermis prior to, during and to after treatment reduces discomfort• minimizes thermal injury to non-target non- skin and skin structures.
Collagen Remodeling & Contraction• Thermally injure a dermal region• Initiates wound healing response• Stimulates fibroblast activity• Denatured collagen leads to contraction• Create collagen remodel or replacement
Treatment Expectations• Expect subtle changes that can vary from immediate improvement to subtle results• Variable Factors –Technique –Patient’s Degree of Laxity Patient s
Sciton SkinTyte Using a controlled thermal profileSkin is cooler at the surface.Its temperature depends on the environment. Room temperature 28°C - 32°CEpidermisDermis 37°C Body temperature
Sciton SkinTyte Chilled Sapphire Plate 12°C Epidermis Dermis 37°C The plate temperature setting BBL s BBL’s Sapphire Chilling plate precisely determines the new thermal gradient fixes the surface temperature.
Sciton SkinTyteChilled Sapphire Plate 12°C Epidermis Dermis 55°C A subsurface layer is briefly heated Then is cooled. cooled heated. SkinTyte elevates subsurface temperature temperature.
Sciton SkinTyte• Denatures collagen causing immediate contraction• Activates collagen generating fibroblasts leading to long-term remodelingEpidermisDermis
Treatment Guidelines• Multiple Passes: Up to 5• Moderate fluences : 40 – 45 J/cm2• Pulse Width: 5 seconds• Cooling : 12°C 12°• Post Cooling: 2 seconds• Treating to a visible, immediate endpoint p• Immediate tightening• Erythema
SkinTyte treatmentPre before Postx’s at 1 month intervals 3 x’3tx Photos courtesy of Jason Pozner, MD
SkinTyte treatment before after 3 monthly treatments Photos courtesy of Jason Pozner, MD