Slideshare uses cookies to improve functionality and performance, and to provide you with relevant advertising. If you continue browsing the site, you agree to the use of cookies on this website. See our User Agreement and Privacy Policy.

Slideshare uses cookies to improve functionality and performance, and to provide you with relevant advertising. If you continue browsing the site, you agree to the use of cookies on this website. See our Privacy Policy and User Agreement for details.

Successfully reported this slideshow.

Like this presentation? Why not share!

- sector-sphere by xlight 1266 views
- Sector Cloudcom Tutorial by lilyco 842 views
- Sphere international 2017 by Niomi Cowling 471 views
- Integrating Sphere by Spectroscopy Shop 132 views
- Volume of cylinders cones and spher... by shumwayc 6307 views
- G:\Panitia\Presentation Earth As A ... by Hamidah Hassan 11645 views

No Downloads

Total views

973

On SlideShare

0

From Embeds

0

Number of Embeds

3

Shares

0

Downloads

23

Comments

0

Likes

1

No embeds

No notes for slide

- 1. Integrating Spheres Jehona Salaj jehonas@student.uef.fi University of Eastern FinlandDepartment of Physics and Mathematics November 6, 2012
- 2. Figure : Sculpture of the integrating sphere in the Technical University ofDresden (photo:Kay K¨rner). o
- 3. Uses of Integrating Spheres Alone or as accessory of other devices In radiometry and photometry For measuring transmittance and reﬂectance For measuring the light sources E , I and Φ
- 4. Outline 1 The Sphere 2 Theory 3 Designing an integrating sphere 4 Measurements
- 5. Outline 1 The Sphere 2 Theory 3 Designing an integrating sphere 4 Measurements
- 6. The sphere Figure : Scheme of an integrating sphere. Note! An integrating sphere spatially integrates the radiant ﬂux.
- 7. Outline 1 The Sphere 2 Theory 3 Designing an integrating sphere 4 Measurements
- 8. Radiation Exchange Figure : Radiation exchange between dA1 and dA2 . cos θ1 cos θ2 A2 A2 dFd1 −d2 = 2 dA2 ⇒ F1−2 = 2 = (1) πS 4πR AS
- 9. Surface radiance Figure : Surface radiance LS for the input ﬂux Φi Φi ρ LS = × (2) πAS 1 − ρ(1 − f )
- 10. The sphere multiplier The second part of equation (2) is the sphere multiplier. Considering the wall reﬂectance as average and the port reﬂectance zero we get: ρ0 M= (3) 1−ρ ¯
- 11. Spacial and temporal integration Integrating spatially: Φ = Φi ρn (1 − f )n (4) Temporal response is of form e −t/τ (5) where 2 DS 1 τ =− (6) 3 c ln ρ ¯
- 12. Outline 1 The Sphere 2 Theory 3 Designing an integrating sphere 4 Measurements
- 13. The Sphere diameter Radiance relates to the sphere diameter: M LS ∝ 2 (7) DS Decreasing port fraction increases M Port fraction ≤ 5% of the sphere surface Note! Best choice: Large sphere diameter and small port size.
- 14. Use of baﬄes Figure : The use of baﬄes in the integrating sphere. Baﬄes help preventing that the direct incident light enters the ﬁeld-of-view of the photodetector.
- 15. Use of diﬀusers Figure : The use of an auxiliary or satellite integrating sphere as a diﬀuser. If the sphere is used as a collector for measuring radiant ﬂux, the error increases if the incident ﬂux enters the detector’s ﬁeld-of-view.
- 16. Detector in use Figure : Use of lens for collecting the light to the active area of the photodetector. Without a lens: Φd = LS Ad π sin2 θ (8) Putting a lens in the system: π Φd = LS Ad ε0 (9) (2f / )2
- 17. Fiber in use Figure : Coupling the light out using an optical ﬁber. Φf = LS Af π(NA)2 (1 − R) (10)
- 18. Choosing sphere coatings Two important factors: Reﬂectance Durability
- 19. Sphere coatings Some usual coatings: barium sulfate based spray coatings packed PTFE coatings Labsphere’s proprietary reﬂectance materials and coatings:
- 20. Sphere coatings(cont.) Spectralon (over 95% reﬂectance at 250nm to 2500nm; stable even above 350◦ C ; durable over 100h under UV ﬂux exposure.) Spectraﬂect (barium sulfate; 98% at 400nm to 1100nm; durable up to 350◦ C ; not good in humid environment; cheap.) Duraﬂect (94 to 96% reﬂectance over 350nm to 1200nm; good in humid environment; not good for UV range uses; not compatible with some plastic substrates.) Infragold (electrochemically plated; gold metallic reﬂectance coating; 92 to 96% reﬂectance over 1µm to greater than 20µm)
- 21. Outline 1 The Sphere 2 Theory 3 Designing an integrating sphere 4 Measurements
- 22. Radiometers and photometers Figure : Use of integrating sphere as a radiometer or photometer: (a)Sphere Photometer, (b)Laser Power meter, (c)Cosine receptor.
- 23. Reﬂectance and transmittance Figure : Measuring reﬂectance and transmittance.
- 24. Measurement geometries ”d/0◦ and 0◦ /d” The geometries used when dealing with integrating spheres are indeed d/8◦ and 8◦ /d, but are considered d/0◦ and 0◦ /d (as everything with an angle smaller than 10◦ ).
- 25. Questions or comments?

No public clipboards found for this slide

Be the first to comment