Be the first to like this
This is to help you get your cochlear implant working with an FM system.
Why? Because we have had difficulty getting Cochlear’s CP810 (Build D) Speech processor and two different FM receivers (Phonak’s MLXi and ML14i) to work together.
We found the MLXi and ML14i have different start up sequences with the CP810.
If we don’t follow these sequences exactly,our daughter who has bilateral (left and right) implants hears no sound from either the microphones or the FM receivers.
If you have constructive comments or more useful information please add that to the SlideShare page of this presentation.
Slides by David Lovell and Rachel Tyson
Huge thanks to:
Anne-Marie Crowe (Sydney Cochlear Implant Centre)
Rachel Middleton (Australian Hearing)
Micah Mann (Australian Hearing)
Peter Wohlfahrt (Phonak)