Siemens siremobil iso c c-arm with lithotripterPresentation Transcript
Siemens Siremobil ISO-C C-Arm with Lithotripter D O M: 2000
Specifications SonolithPraktis Lithotripter - Serial Number SP027 Sony UP-960CE Printer and footswitch
Report C-Arm Powers up and self tests ok. Both monitor and control console working ok. Function keys and numeric display in good working order. Footswitch and handheld exposure switch functional. Exposure initiates and good image quality obtained. B/W graphic printer in good working order. Aiming laser fully functional. C-Arm in very good condition at functional at time of test. I.I 9 inches.
Report Lithotripter powers up and self test ok. Hand held control display and soft key fully functional. Shockwave generator setting and power control in good working order. Test carried out with test cylinder and capsule, all systems in good working order. Shock wave delivered and capsule disintegrated. Table attachment fully operational with all motorised positions working ok. The electrode storage bench comes with unit but no vacuum pump. One new electrode, sealed but expired January 2010. Plus one used with 4499 pulses delivered. Also include US localisation test tool system in very good cosmetic condition and functional at time of test.